1 article(s) from Schumacher, Zeno
Q-factor of NCLR cantilever with different coating coverage percentages. A 30 nm Al coating was add...
Jump to Figure 1
Noise spectra for soft cantilever with different coating coverage acquired in air. Fully coated can...
Jump to Figure 2
(a) Force noise density spectra for the soft cantilevers obtained from Figure 2 by multiplying with the mea...
Jump to Figure 3
Force–distance curves for an uncoated (red), partial coated (blue) and fully coated soft cantilever...
Jump to Figure 4
Calculation of the change of Q-factor for a fully coated NCLR cantilever with different coating thi...
Jump to Figure 5
Beilstein J. Nanotechnol. 2015, 6, 1450–1456, doi:10.3762/bjnano.6.150
Subscribe to our Latest Articles RSS Feed.
Register and get informed about new articles.
Follow the Beilstein-Institut