1 article(s) from Ahmed, Zuned
CAD drawings of the top and side views of the UHV system consisting of a cryostat chamber, a prepar...
Jump to Figure 1
(a) The bath cryostat consists of two tanks, the inner tank holds 8 L of LHe and the outer tank hol...
Jump to Figure 2
Schematics of the components of (a) a typical, classical STM/tuning fork-based AFM setup with the t...
Jump to Figure 3
CAD drawings of the cylindrical body tube (a) and the leaf spring (b) carrying two of the total of ...
Jump to Figure 4
(a) CAD drawing of the z-positioning unit containing the xy-positioning units with scan piezo and, ...
Jump to Figure 5
CAD sketch of the fiber z-positioning unit containing the x- and y-positioning unit. The assembly c...
Jump to Figure 6
(a, b) Top and side view CAD sketches of a sample holder with a button heater for sample preparatio...
Jump to Figure 7
(a) Setup of the interferometer system. (b, c) Amplitude and phase as a function of the frequency f...
Jump to Figure 8
(a) Narrow band thermal noise spectrum of a NCHPt cantilever with length L = 125 μm and width w = 3...
Jump to Figure 9
The current noise spectral density with the tip retracted from and approached to the surface of an ...
Jump to Figure 10
(a, c) STM results on Au(111) obtained at 600 mV and 20 pA. (b, d, e) Cross sections taken at the l...
Jump to Figure 11
AFM data: (a) 400 × 400 nm2 image of NaCl islands on a Au(111) surface scanned with the cantilever ...
Jump to Figure 12
Multichannel and multimodal AFM results obtained on a NaCl island running over a Au(111) step edge....
Jump to Figure 13
Beilstein J. Nanotechnol. 2022, 13, 1120–1140, doi:10.3762/bjnano.13.95
Subscribe to our Latest Articles RSS Feed.
Register and get informed about new articles.
Follow the Beilstein-Institut