2 article(s) from Petersen, Dirch H.
The standard probe pin configurations A, A’, B and B’ used in the experiments.
Jump to Figure 1
The three sub-probes on an M7PP used for multiplexing during measurements; a) 1357 (20 μm pitch), b...
Jump to Figure 2
The dashed curves show the relative Hall signal ΔRBB′/RH = f(ζ) (Equation 1) and relative pseudo sheet resist...
Jump to Figure 3
Static position errors: at the top, the ideal positions of a M7PP is shown. Below, the case of only...
Jump to Figure 4
Effective sensitivity for a) R0, b) NHS and c) μH when in- and off-line errors are present during ...
Jump to Figure 5
Effective sensitivity for the sheet resistance, Hall mobility and Hall sheet carrier density, due ...
Jump to Figure 6
Generalization of Figure 6. The sensitivity of each parameter, a) Sheet resistance R0, b) Sheet carrier den...
Jump to Figure 7
Beilstein J. Nanotechnol. 2018, 9, 2032–2039, doi:10.3762/bjnano.9.192
Top-view schematic of the four μ4pp electrodes landed on (a) a single fin and (b) two fins. The ele...
(a) Measured fin resistance Rfin as a function of fin width Wfin on isolated (triangle) and dense (...
(a) TEM image of four ca. 20 nm wide Si fins where the measured Rfin is indicated on top of each fi...
Beilstein J. Nanotechnol. 2018, 9, 1863–1867, doi:10.3762/bjnano.9.178
Subscribe to our Latest Articles RSS Feed.
Register and get informed about new articles.
Follow the Beilstein-Institut